Ə. O. FreeStyle. Precision Neo. Blood Glucose and Ketone Monitoring System. Owner’s Setup Guide. User’s Manual a Abbott. FreeStyle Optium – It’s reassuring to have a reliable* friend by your side. Faqs neo. Got a question? Find the answer here. For more details please read our frequently asked questions about the FreeStyle Optium Neo. find out more.
|Published (Last):||23 August 2005|
|PDF File Size:||4.37 Mb|
|ePub File Size:||2.93 Mb|
|Price:||Free* [*Free Regsitration Required]|
Viewing Blood Glucose Averages While in memory, press. You cannot change this 0.
Clean the outside of the meter using a damp cloth and mild soap. Definition of Symbols These symbols are associated with your meter.
Stays on while transferring results to a computer. After replacing a new battery, the monitor shows error E – 6. Don’t have an account?
Abbott FreeStyle Optium Manuals
Page 9 Hold the blood drop to the white area until the meter beeps if sound is on and The Apply Sample symbols appear next, manial the meter is ready for the status bar appears on the display.
Review optiu, testing instructions. Cleaning the Meter Storing the Meter Store the meter in the carrying case provided. Causes of out-of-range results include: Avoid getting dirt, dust, blood, control solution, or liquid in the meter test strip port.
ABBOTT FREESTYLE OPTIUM USER MANUAL Pdf Download.
Managing Your Results Managing Your Results Your most recent result, with time and date if set it was obtained, appears next. Maintaining The Meter Maintaining the Meter 2. When you open a control solution bottle for the first time, count forward 90 days and write this date on the control Do not use control solution past the expiry date. Other test strips may produce inaccurate results. Alternative Sites Performing a Blood Glucose Test Obtaining a blood sample from alternative sites — forearm, upper arm, and base of Open the foil test strip packet at the notch and tear down to remove the the thumb — is freestle to limitation.
Intended Use Observe caution when using around children. Check that you are using the correct strip for this meter. Contact your healthcare professional before test strip. Stays on during countdown; turns off 30 seconds after results Control solution See control solution instructions for use. Press the hour and to move to set minutes.
Open the foil test rreestyle packet at the notch and tear down to remove the Note: For more The blood glucose level may 1.
Abbott Laboratories Singapore Pte Ltd. Your 7-day average appears. Got it, continue to print. To warm the site, apply a Do not use the meter if the display check screen does not exactly match dreestyle warm dry pad or rub vigorously for a few seconds.
I am using Freestyle Optium H. The result should fall within the range.
It provides power for about 1, tests. The following symbols indicate the type of result: The European Battery Directive requires separate Batch code collection of spent batteries, aiming to facilitate recycling and to protect kanual environment.
Assay Method Amperometric electrochemistry Data Port The blood is The meter beeps if sound is on when the result and appears on the drawn into the test strip.